quality stethoscope, that has been fitted with the adapter, which
adds an electronic sensor to the acoustic tube. Allows examiner to
listen acoustically while concurrently recording and displaying heart
or lung sounds on the Handheld STG or STG for PC. The
STG Stethoscope™ cannot be sold as an independent product.
It is only available as part of a system, or a replacement part.
To purchase, please email email@example.com
Tap Start > Settings > System Tab > iPAQ Audio > Microphone
2. Disable Automatic Gain Control.
3. Set Microphone Gain to 27.
4. Tap OK.
5. Tap Start > Settings > System Tab > Backlight > Battery
6. Adjust 'Turn of backlight if device is not used for' to 3
Connecting STG Stethoscope to iPAQ:
1. Start recording with the STG software
2. Gently tap the diaphragm. If you don't see sharp spikes occupying
all vertical space of the screen, disconnect and reconnect iPAQ cable
from the iPAQ until you see spikes. The spikes will appear when iPAQ
switches to record from external microphone.
3. Firmly position stethoscope and start recording.
||The sound that you hear with the STG Stethoscope
is the sound that you hear with any acoustic stethoscope.
||The STG Stethoscope acoustic quality is comparable
to Littmann Master Classic II.
|Bell and Diaphragm Modes
||The chest piece lets you monitor low and high-frequency
sounds by alternating pressure on the chest piece. Reduce pressure
for the bell mode, increase pressure for the diaphragm mode.
|Chest piece rim
||Nonchill plastic rim
||Soft sealing ear tips
Recording and Visualization
|System Frequency Response
||50Hz to 2000Hz
||1.2V Min, 20V Max
|Rated Current Drain
||130 µA Max @ Ta –25°C
||-40°C to +70°C
||Electret microphone is positioned in the space continuous
with the tubing chamber.
||3.5mm stereo audio jack. signal is on ring and tip
that are tied together.
|Safety and EMI tests
||Has passed Safety and EMI tests.
Holding a chest piece by hand often generate artifacts that make visual
display unreadable. If you have this problem (and most people do), we
suggest that you attach the chest piece to skin so that you do not need
to hold it. There are several attachment methods you can use:
1. Just place the chest piece on bare skin. If the chest piece is heavy
enough it will stay in place.
2. Attach the chest piece with 3M Double Stick Discs (SKU B11687). You
can order it here: 3M
Double-Stick Discs pack of 102 Size 1 1/4" 3M SAS2349 (HC) 3M 2181 (Pack).
3. Use a rubber belt to hold chest piece in place. We prefer maternity
belt used to hold a doppler monitor in place when the hospitals want to
hear the fetal heartbeat for an extended time. It's made of very soft,
pliable and stretchy material. You can order it here: www.utahmed.com,
Toco Abdominal Belts, Model ABC-4425